Thermal deformation testing machine
Product origin :China
Delivery time :10 days
Supply capacity :1000
Temperature control range: environmental temperature - 300degree
The heating rate: (120±10)degree/h or(50±5)degree/h
Maximum deformation error: ±0.005mm
Temperature control error: ±0.5degree
Show precision: ±0.001mm
Deformation measurement range: 0-1.5mm
Distortion Range: 0--1.5mm
Sample frame: 3
Heating Medium: methyl silicon oil ：3kw
Power Supply: AC220V 50Hz
Cooling method: 150 degrees Celsius above natural cooling or air; 150 degrees below natural cooling or air.
Thermal deformation testing machine, vicat softening point temperature instrument using single-chip microcomputer and tester, adopts LCD processing temperature and deformation, set arbitrary sample deformation (deflection value), the rate of temperature, temperature test parameters, etc. on the control panel. Thermal deformation testing machine is simple and easily uses, good performance, high precision, mild products with test data storage function, and in the test process can control temperature and deformation, At the end, the system alarm automatically and then stop heating and memory test data.
Thermal deformation testing machine, Vicat softening point temperature instrument complies with ISO75 (E), ISO306 (E), GB/T8802, GB/T1633, and the standard of GB/T1634 etc. The thermal deformation of polymer material, Vicat softening point temperature measurement. Thermal deformation testing machine points A and B type two kinds, type A for the desktop Type B for vertical, Type B experiment with automatic lift-and-descend function.
1. Open the machines everyday periodically.
2. Operated by the specialist.
3. Thermal deformation testing machine and accessories must be deposit in the dry environment.
4. After testing the accessories should be added wiping grease rust, and remove the rust great before test.
5. When the test finished, the power should be closed to avoid any accident.