High and low temperature alternation temperature humidity test chamber
Product origin :China
Delivery time :10 days
Supply capacity :1000
1. Type: GDJS-100
2. Inner size W*H*D(mm): 350*350*450
3. Temperature range: -40C~+150C
4. Temperature fluctuation: ±0.5C
5. Temperature uniformity: ±1C(Empty load when +80C)
±2C(Empty load when -40C)
6. Average rate of raise and drop the temperature: raise 1~3C/min, drop 0.7~1C/min
High and low temperature alternation temperature humidity test chamber is suitable for industrial and electronic products, performance reliability test, this test has a greater temperature control, performance indicators to and meet GB10592-93《high-temperature chamber technical conditions》, GB10586-93 《heat chamber technology》also can be used for GB2423-93《electric and electronic products environmental testing procedures》, 《low-temperature test A test methods; test temperature B test methods》 and the constant heat, damp heat test, High and low temperature alternation temperature humidity test chamber can be effective low-temperature and high humidity test.
1. Compress machine: The original French "tecumseh" closed compressor set.
2. Condenser: The air-filled type plate condenser
3. Evaporimeter: Fins chip sections.
4. Accessory: imported refrigeration desiccant, refrigerant flow inspection, the United States the oil-water separator.
5. Swell system: Italy swell valve.
1. China and English menu mode.
2. R232 communication interface( with software)
3. Curve schemes displayer.
4. With a function of subscribe start-up and shutdown.
5. With a function of automatic memory when it is outage.
6. High and low temperature alternation temperature humidity test chamber can be set by program or pointed.
7. Accuracy: 0.01C
8. Resolution: 0.01C
9. PID parameter control, maximum 10 program memory, 100section, and can repeat running 200 times per single program.
10. Senor: PT100 senor.
1. Open the machines everyday periodically.
2. Operated by the specialist.
3. High and low temperature alternation temperature humidity test chamber and accessories must be deposit in the dry environment.
4. After testing the accessories should be added wiping grease rust, and remove the rust great before test.
5. When the test finished, the power should be closed to avoid any accident.