Level and direct combustion flame retardant analyzer
Product origin :China
Delivery time :10 days
Supply capacity :1000
Supply voltage and power：220V±10%，50HZ
Power: less than 100W
Automatically ignited time of the Bunsen burner：10s 12s 30s
Ram of the Bunsen burner: 9.5mm±0.5mm
There is sound tips of the Bunsen burner at the time it finishes impose flame to the sample.
Level and direct combustion flame retardant analyzer has the function of recording and storing the burning time of flame and no flame of the samples and calculating the speed automatically.
Gas source：liquefied petroleum gas
Instrument weight：about 40 Kg
Instrument size: about：1100 mm×600 mm×1350 mm
The size of the sample:
|Level method||125±5||13.0±0.3||3.0±0.2||3 rose|
|Vertical method||125±5||13.0±0.3||3.0±0.2||5 rose|
On the horizontal and vertical direction with small sample placed flaming fire point and then determined the flame burning is burn and no flame combustion time and burning length and burning rate.
Level and direct combustion flame retardant analyzer was developed on the basis of the national standard, GB/T2408-2008 / IEC60659-11-10:"Determination of the level of plastic combustion in horizontal and vertical method" of 1999. Level and direct combustion flame retardant analyzer completely meets the technical requirement of the national standard GB/T2408-2008 and UL-94. The plastic performance determination in level method and vertical method can be toke in the same machine. Automatically record the time of test and permit the user to input the length of over firing in the level method to calculate out the speed of burning automatically. Operated by computer and showed on the computer. Level and direct combustion flame retardant analyzer can be use both in windows XP and windows Vista operating system, and the test report can be printed by the printer.(This instrument include type CZF-5 and CZF-6.)
1. Open the machines everyday periodically.
2. Operated by the specialist.
3. Level and direct combustion flame retardant analyzer and accessories must be deposit in the dry environment.
4. After testing the accessories should be added wiping grease rust, and remove the rust great before test.
5. When the test finished, the power should be closed to avoid any accident.